Errata and Updates
The station scientists are responsible for the quality of their submitted data. Within the archive all incoming data are format-checked and visualized. Files with obvious errors are not imported into the archive.
Nevertheless, errors within the archive cannot be fully excluded. If errors have been detected the erroneous data are replaced by corrected datasets. If corrected data cannot be submitted within a reasonable period of time, the questionable data will be deleted in PANGAEA, and a warning will be given in the corresponding ftp files.
The version of the data is given in the first line of the station-to-archive files. New versions within PANGAEA obtain new doi-numbers. The old doi-numbers will not be deleted but linked to the new version. A comment in the PANGAEA meta-data displays the version number and the date of the change if more than one version has been submitted.
This page is written to inform the customers about detected errors and updated versions.
Currently the files from PTR,RLM,BRB,SMS until 2008-08 still contain wrong long-wave fluxes. The same error exists in the long-wave data from FLO after August 1999. The wrong long-wave fluxes have been deleted in PANGAEA. It might be, that the data declared LWD are actually LWU fluxes.
The corresponding station-to-archive files are no longer readable for BSRN-users. In case you need a copy nevertheless, please contact: Amelie Driemel. The files will be replaced as soon as updates are available.
In BRB there seems to have been a tracker problem in 2014: the data from April until August as well as some from October show too low SumSW values compared with SWD at around noon and too high values in the afternoon, see also http://bsrn.awi.de/products/quality-code/comparisons/brasilia-brb/.
2017-05-17: PAY LR3030
Between 1996 and 2004 the SWD data in logical record LR3030 seems to be wrong (values too low). The station scientist is informed, he already confirmed that the data seems to have been mixed up. The files will hopefully be replaced soon.
The station scientist informed us that at station Tiksi, wrong calibration coefficients have probably been used to obtain the data submitted to BSRN. This will be looked into soon, and data resulting from wrong calibration coefficients will be replaced.
On 3 November 2011 there has been a change of instruments used for the measurement of the downward longwave irradiance (DL in LR0100). This change of instruments was accompanied by a calibration change, removing a bias of +3.85 ± 1.48 W/m2 that exists in DL before 3 November 2011. After 3 November 2011, the values of DL are traceable to the WISG. We plan to correct the data (remove the bias) before 3 November 2011 and resubmit the station-to-archive files (February 2005 – November 2011).
For further information please contact Wouter Knap.
2014-12-11: E13 error in LR3010
In logical record LR3010 of station E13 from January 1994 to June 2009 (i.e. files e130194.dat.gz to e130609.dat.gz) the values of LWU and SWU are accidentally interchanged. The problem has been solved for the files available in PANGAEA. However, the ftp files still need to be replaced!
Some additional quality control is needed for the LW data collected at Ilorin. There are issues related to the way the measurements were taken (shaded, not shaded) and it has to be accounted for. Rachel T. Pinker hopes to do it this summer. For further information please contact: firstname.lastname@example.org.