Errata and Updates
The station scientists are responsible for the quality of their submitted data. Within the archive all incoming data are format-checked and visualized. Files with obvious errors are not imported into the archive.
Nevertheless, errors within the archive cannot be fully excluded. If errors have been detected the erroneous data are replaced by corrected datasets. If corrected data cannot be submitted within a reasonable period of time, the questionable data will be deleted in PANGAEA, and a warning will be given in the corresponding ftp files.
The version of the data is given in the first line of the station-to-archive files. New versions within PANGAEA obtain new doi-numbers. The old doi-numbers will not be deleted but linked to the new version. A comment in the PANGAEA meta-data displays the version number and the date of the change if more than one version has been submitted.
This page is written to inform the customers about detected errors and updated versions, or about stations behind schedule and the reasons hereof.
2022-02-01 Granite Island (GIM) data loss
The station got hit by an ice storm at the beginning of January 2022, and everything got covered with a layer of ice. Also, there was such a long period of heavy overcast that the solar power system failed. So there is no data at all for the rest of the month. The station will probably not send any data until May/June 2022.
Granite Island (GIM) data collection is back up and running. Batteries and tracker were replaced and newly calibrated Shortwave and Longwave instruments as well as other hardware/instrumentation were (re)installed at the site.
2021-12-20 Magurele station info
Please note, that the SW data for our newest BSRN station Magurele (INO) are preliminary. Although generally of good quality, there is a discrepancy between SWD and SumSWD during sun conditions, so most probably a new version with new calibration will be uploaded in summer 2022.
2021-09-24: LIN behind schedule
Station Lindenberg ist behind schedule due to reconstruction issues and data acquisition exchange. The plan is, however, to resume submissions in November 2021.
2021-09-20: CAM and LER behind schedule
Stations Camborne and Lerwick are behind schedule due to staff turnover and a subsequent effort to refurbish\replace existing systems at both sites. The data submission will hopefully be back in line in mid-2022.
2021-09-15: BAR, BER, SPO behind schedule
For stations Barrow, Bermuda, and South Pole, the data processing is behind schedule due to staff turnover. We expect to be caught up by Summer of 2022.
2021-01-01 (ongoing problem): No DIR and DIF data at TAM since 2018
Due to a broken tracker which could not yet be replaced the DIR and DIF data at Tamanrasset is missing since February 2018.
2020-01-01 (ongoing): Ny-Ålesund (NYA) LR1100
For station NYA the LR1100 record ends in 2017. However, high resolution radiosonde measurements from 2017 on can be found here.
2020-01-01 (ongoing problem): PTR,RLM,BRB,SMS
Currently the files from PTR,RLM,BRB,SMS until 2008-08 still contain wrong long-wave fluxes. The same error exists in the long-wave data from FLO after August 1999. The wrong long-wave fluxes have been deleted in PANGAEA. It might be, that the data declared LWD are actually LWU fluxes.
The corresponding station-to-archive files are no longer readable for BSRN-users. In case you need a copy nevertheless, please contact: Amelie Driemel. The files will be replaced as soon as updates are available.
In BRB there seems to have been a tracker problem in 2014: the data from April until August as well as some from October show too low SumSW values compared with SWD at around noon and too high values in the afternoon, see also http://bsrn.awi.de/products/quality-code/comparisons/brasilia-brb/.
2019-08-21: BSRN Toolbox errors in "Expanded measurements.." LR1300
In August 2019 it was noted, that the Toolbox code was giving wrong values in the extraction of LR1300 (https://dataportals.pangaea.de/bsrn/?q=LR1300). If the station-to-archive file contains a value of 10000 in the parameter cloud base height then the Toolbox gives out a value of 99999 which means "no cloud". This issue has to be looked into, please be careful when using the data.
2017-10-01: Barrow files fixed
2017-05-17: PAY LR3030
For further information please contact Wouter Knap.
Some additional quality control is needed for the LW data collected at Ilorin. There are issues related to the way the measurements were taken (shaded, not shaded) and it has to be accounted for.